For more info: 021 55734344
Four measuring sections with different thermometers can be individually selected using ball valves. The temperatures measured using the resistance thermometer and the thermocouple are displayed digitally. The bimetallic dial and liquid-in-glass thermometers are direct indicating instruments.
The response of the measuring devices for different flows can be investigated. The flow rate is measured with a rotameter. The effect of different installation positions and methods on measurement accuracy can be investigated.
Learning Objectives / Experiments
- Function, type of construction and applications of bimetallic dial thermometers
- Function, type of construction and applications of liquid expansion thermometers, resistance thermometers and thermocouples
- Measuring precision, sensitivity and measuring errors of the different thermometers
- Installation methods, installation errors and response